Add: 4th Floor, Area C, Building 1, Aishang Technology Industrial Park, No. 108, Honghu Road, Songgang Street, Baoan District, Shenzhen, Guangdong Province
Sometimes when using the battery cell activation instrument, you will encounter some problems. Here are the solutions to several common problems during use:
1. No response after booting
The power is not connected. The battery pack load tester has a built-in memory in the host computer. U disk can be used to read and store records. You don’t need to carry a computer. It can work on a stand-alone basis. In the future, you can print the report with the computer. It can have the voltage and current calibration correction function, and the measured value of the meter can be carried out at any time. Calibration correction to ensure measurement accuracy. Please check whether the power plug of the battery cell activation device is properly inserted. The terminal wire should be pressed tightly. The host menu operation control is normal, and the external signal has no response
The host has poor contact (loose wiring). Please check whether the contact is reliable. Key failure or confusion
Please check if there is a key stuck but not popped up. If there is, make it pop up to resume normal operation. Use activation to perform N times of separate charging and separate discharge
Because the charging and discharging time alone cannot exceed 18 hours, if the charging and discharging time alone is required to exceed 18 hours, use the activation N cycles (continue) to increase the time. Storage battery internal resistance tester The instrument online test record can be displayed and the voltage, resistance, capacity or other important parameters of a single cell is effectively pick behind the battery, the battery and the computer-specific data management software and Generate test reports, track battery decay trends, and provide maintenance recommendations. It is suitable for maintenance and inspection of storage batteries in communication base stations, substations and UPS. Used for battery acceptance, battery grouping and routine inspection. When using the battery tester , put the nickel-metal hydride battery or lithium battery to be tested into the tester, adjust the various indicators, and perform the test, and then the battery voltage, internal resistance, capacity, charge, discharge, and overcurrent can be tested. , Short circuit protection time. After the test is completed, press the "print" button, and a small printer can print out a test report similar to a taxi ticket. The report can display the battery type, voltage, capacity, test unit, time and eligibility. Example: If the charging time is 28 hours, set the number of activation cycles to two, the first charging time is 18 hours, the discharge time is 0, the second charging time is 10 hours, and the discharge time is 0. The discharging method is the same as above, and the charging time should be 0 at this time.
Battery number setting
The battery number input bit is 4 digits. To facilitate management, the first two digits are set as the group number, and the last two digits are set as the battery number. There can be a maximum of 99 groups, and the maximum number of batteries in the same group is 99. Error writing to U disk
1. Check whether the U disk is inserted properly;
2. See if the attribute of Disk 0 is in FAT format, if not, please format the U disk with FAT;
3. If there is an error or no response for a long time during the process of reading and writing the U disk, please format the U disk with FAT and try again.
Contact: Mr. Luo
Address: 4th Floor, Area C, Building 1, Aishang Technology Industrial Park, No. 108, Honghu Road, Songgang Street, Baoan District, Shenzhen, Guangdong Province